FLUKE 80T-IR PDF
80T-IR – read user manual online or download in PDF format. Pages in total: Find great deals for Fluke 80t-ir Temperature Probe 80TIR. Shop with confidence on eBay!. Find great deals for Fluke Infrared Temperature Probe M# 80t-ir. Shop with confidence on eBay!.
|Published (Last):||11 December 2016|
|PDF File Size:||7.37 Mb|
|ePub File Size:||4.67 Mb|
|Price:||Free* [*Free Regsitration Required]|
Verification Points Model Blackbody Temperature at 0. Sending feedback, please wait Search the history of over billion web pages on the Internet. May substitute less accurate blackbody if using in conjunction with transfer standard of equivalent accuracy, i. Save to parts list Save to parts list. You agree that the Information as provided here through an intermediary may not be error-free, accurate or up-to-date and that it is not advice. Accuracy figures are valid for a period of one year.
Recommended Equipment Table 1 lists the equipment required to perform the calibration verification procedure. Refer to Figure 1 for equipment configuration.
In between measurements move the UUT to an environment not affected by the heat source. Replace the battery if necessary. The product does not contain any of the fluie substances in concentrations and applications banned by the Directive, and for components, the product is capable of being worked on at the higher temperatures required by lead—free soldering.
Fluke 80T-IR Non-Contact Temperature Probe
The Manufacturers disclaim all warranties including implied warranties of merchantability or fitness for a particular purpose and are not liable for any damages arising from your use of or your inability to use the Information downloaded from this website. Reactivate by switching the UUT off and on again. Save to an existing parts list Save to a new parts list. Record the reading displayed on the DMM. Add to a parts list. Repeat steps 1 through 5 for the remaining verification points called out in Table 2.
Please select an existing parts list.
Fluke Infrared Temperature Probe M# 80t-ir | eBay
Power up the UUT and allow it to stabilize for five minutes. You have chosen to save the following item to a parts list:. The foregoing information relates to product sold on, or after, the date shown below.
If your instrument fails to pass any performance specifications, return it to the Eluke Eactory or Service Center for repair. The product detailed below complies with the specifications published by RS Components.
Fluke 80T-IR Infrared Temperature Probe
Thank you for your feedback. Place the UUT aperature three inches from the center and perpendicular to both axes of the Blackbody plate.
Set the DMM to the 3. Please enter a message. Adjust the setpoint of the Blackbody to output each of the temperatures in Table 2 at 0.
The Manufacturers reserve the right to change this Information at any time without notice. With the UUT powered on, check that the battery voltage is at least 7. Save this item to a new 80t-ig list.